Asthma Inhaler Mini Automizer For Children Adult Inhale Nebulizer Ultrasonic Nebulizer Spray Aromatherapy Steamer Health Care
Two airflow control,
Compact and portable,
Suitable for all ages people.
UK adapter included(Note: the default plug is UK standard. We will provide free conversal adapter for all buyers according to your location/area and demand. )
The better way use about 20-60 degrees hot water pour into big cup when you first time use it, only can be use clear water.
The machine itself does not have the heating function.
1. HKz: 2.4MHz
2. Particle size: Res pirable Fraction 0.5
3. Maximum Med capacity:15ml
4. Machine Capacity: 25ml
5. Average Nebulization Rate: 0.4ml/min
6. Working voltage: 12V DC
7. Power Consumption: 6w
8. Weight: appr. 0.5KG
9. Input voltage: 110V-240V
1. Remove the top cover, take the cup and silicon gasket out.
2. Add 30ml purified water into the Water Chamber, the water level should be 3-5mm above the external silicon ring.
3. Fill the medicine cup half-full. put the cup into the chamber.
4. Re-install the top cover securely with the silicon gasket.
(Note: the best atomisation temperature in the chamber is 30℃/8℉)
Please use saline water or WARM(NOT boiling) water to sterilize the accessories.
Only the cup with water can the machine work, otherwise it will damage the machine, the cup under the medicine cup must be added water!Any question please contact us before open dispute or leave us bad feedback.
1. Pease add purified water instead of running water into the water chamber.
2. The machine should be cleaned promptly after use.
3. Please check whether the machine is damaged, cracks, est. before and after using the product. If yes, please contact our customer service for replacements.
1.If the machine doesn’t work or red light flashes, check whether there’s enough water in the water chamber, ensure that the Atomizing Cup can touch the water surface.
2. If completely no mist or small mist, check the following:
a. If the water tempreature is below 20℃/68℉, replace it with water at 40℃/104℉.
b. If the medicine liquid has a high concentration or small quantity, it needed to be diluted with normal saline before operate.
c. If the liquid for atomization contains large particles/impurity/precipitat/flatage/colloidal particles, the machine would not work.
d. Change a NEW cup, if no mist after few uses.
3.If the top cover does not fit, please ensure the silicon gasket been placed tightly.
4.For water leakage issue, please don’t make the machine tilt.
Ultrasonic frequency: 2.4Mhz; Operation voltage: DC12V
Size of Atomized Particle: 0.5-5 micron
Cup capacity: 10ml; Vessel Capacity: 25ml
Pakage Include( in original packed):
1 x Portable Nebulizer
2 x Mask (1 adult size, 1 children size)
5 x Medical cups
2 x Mouth pieces
1 x AC adapter portable nebulizer
1 x English Manual